Spectroscopy and Optics: Laser mirrors: High reflectance is measured best with cavity ring-down spectroscopy
For mirrors with reflectance values above 99.5%, measuring total loss through cavity ring-down spectroscopy is more accurate than conventional methods.
For mirrors with reflectance values above 99.5%, measuring total loss through cavity ring-down spectroscopy (CRDS) is more accurate than conventional methods.
IAN STEVENSON and CORY BOONE
High-reflectance mirrors, with reflectance values ranging from 99.8% up to 99.999%, are essential components in most laser systems for beam steering while maximizing throughput. It is common industry practice to determine mirror reflectance by measuring transmission using spectrophotometry and assuming that the rest of the light was reflected. However, this false assumption does not consider scatter or absorption, leading to overly optimistic reflectance values. For mirrors with reflectance values above 99.5%, a more accurate way to determine reflectance is to measure total loss through cavity ring-down spectroscopy (CRDS). Understanding your supplier’s metrology is critical for predicting real-world performance.
In many laser applications, including medical laser systems and laser materials processing, there are typically anywhere from 2 to 12 mirrors used for beam steering. Minimal loss and high reflectance values of these mirrors are crucial for maximizing throughput and avoiding system damage, especially when many mirrors are used. Dielectric coatings are typically used to maximize the reflectance and laser-induced damage threshold (LIDT) across a small bandwidth, with typical values for reflectance and LIDT of >99.5% and >20 J/cm2 for 20 ns, 1064 nm pulses. However, for particularly sensitive applications where maximizing throughput is especially important, reflectance values up to 99.999% may be required. These ultrahigh-reflectance mirror coatings are often deposited through tightly controlled processes such as ion-beam sputtering (IBS).
FIGURE 1. Shown is an example of laser-induced damage on the surface of an optic imaged using differential interference contrast (DIC) microscopy.
Any light not reflected will be either transmitted through the system, absorbed, or scattered. Undesired transmission can lead to a beam focusing somewhere else in the system, which may lead to safety issues or laser-induced damage. Absorption is even more likely to lead to laser-induced damage through thermal effects (see Fig. 1). Although scattered light is less focused and intense than the original beam, it can reduce throughput, introduce noise into the system, and potentially lead to damage when using high-power lasers. Unwanted transmission, absorption, and scatter not only impact the final performance of reflective optical components, but they also affect optical suppliers’ ability to determine if the desired reflectance is being met.
FIGURE 2. CRDS is an ideal technique for determining the reflectance of laser mirrors because it accurately measures the total loss of the mirror due to transmission, absorption, and scatter.
Ace in the hole: cavity ring-down spectroscopy
Cavity ring-down spectroscopy (CRDS) provides an ideal solution for determining the reflectance of highly reflective optics by measuring the total loss of the optic, which includes absorption, transmission, and scatter. The reflectance of the optic can then be determined by subtracting the measured loss from 100%.
In CRDS, a resonant cavity is formed by two highly reflective mirrors with reflectance values of around 99.99%. A laser pulse is then sent into the cavity where it oscillates between the mirrors, losing a small amount of intensity at each reflection due to transmission, absorption, and scattering (see Fig.2). An intensity detector is placed after one of the mirrors to record the decreasing intensity of the reflected light as a function of time. This allows the total loss of both mirrors to be determined by the decay time, or “ring down,” of the reflected light inside of the cavity.
The sensitivity of directly measuring a laser mirror’s reflectance with an uncertainty of ±0.1% (left two examples) is two orders of magnitude greater than directly measuring the mirror’s loss with an uncertainty of ±10% through CRDS (right two examples), demonstrating that loss measurements are much more accurate than reflectance measurements for highly reflective optical components.
The intensity of the laser pulse inside of the resonant cavity is described by:
I(t): Intensity of the laser pulse as a function of time
I0: initial laser pulse intensity
T: total cavity mirror loss
c: speed of light
L: length of cavity
The loss value measured in a CRDS system is the loss of both mirrors in the cavity. Therefore, three tests must be performed using reference mirrors to determine the loss of one unknown mirror. Two known reference mirrors are used to form the resonant cavity for an initial calibration measurement (A), and then two more measurements are taken with the unknown test mirror replacing each one of the reference mirrors. Consider the measurement done with the first reference mirror replaced by the test mirror as measurement B and a measurement with the other reference mirror replaced by the test mirror as measurement C. Now, measurements A, B, and C can be used to determine the loss of the unknown test mirror by following the following set of equations where M1 and M2 are the loss of the two reference mirrors and M3 is the loss of the test mirror:
FIGURE 3. When measured using transmission spectrophotometry, it appears that Mirror 2 has a higher reflectance than Mirror 1.
However, when CRDS was used to evaluate the loss of both mirrors, Mirror 1 was confirmed to have reflectance >99.9%, where the reflectance of Mirror 2 was so low that it could not even be measured in the CRDS system. Using reflectance spectrophotometry to measure the reflectance of Mirror 2 in comparison to Mirror 1 revealed that it has much worse performance and a trailing reflectance that falls to 0.5% less than that of Mirror 1, which occurs because of scattering and absorption (see Fig. 4). The reduced reflectance at shorter wavelengths is characteristic of scatter and absorption. Mirror 2 could result in system failure if it was believed to have a reflectance of 99.99% while it truly had a reflectance of 99.5%. Discrepancies between design values and true reflectance values can lead to reduced throughput, safety issues, and possibly catastrophic system damage. Both of the examples above show the importance of using the proper metrology for highly reflective optical components.
FIGURE 4. Using CRDS for Mirror 1 and reflection spectrophotometry for Mirror 2 reveals that Mirror 2 actually has significantly lower reflectance than Mirror 1, contradicting the results of the initial transmission spectrophotometry measurement.
While reflectance may seem like a rather simple specification, it is actually a difficult value to measure for highly reflective optics such as laser mirrors. The best technique for reflectance characterization is dependent on the application requirements and reflectance level of the component. Cavity ring-down spectroscopy (CRDS) calculates high reflectance values by directly measuring the total loss from absorption, transmission, and scatter, and is ideal for quantifying high reflectance values above 99.5%. However, CRDS cannot determine reflectance values below 99.5%, and direct-reflectance measurements through reflection spectrophotometry should be used in these cases. Truly understanding the way that your highly reflective optics have been measured is imperative for predicting how they will perform in your application.
Ian Stevenson is principal engineer and Cory Boone is lead technical marketing engineer, both at Edmund Optics, Barrington, NJ; e-mails: email@example.com and firstname.lastname@example.org; www.edmundoptics.com.